Portable Oximeter and Heart Rate Monitor with 3.5 Inch Touchscreen PM-60A.
Principle of the PM60A patient Monitor is as follows:
Photoelectric Oxyhemoglobin Inspection Technology is adopted in accordance with Capacity Pulse Scanning & Recording Technology,the Pulse Oximeter can be used in measuring the pulse oxygen saturation and pulse rate through finger.The product is suitable for being used in family, hospital, oxygen bar, community healthcare, physical care in sports (It can be used before or after doing sports, and it is not recommended to use the device during the process of having sport) and etc.
Small in volumelight in weight and convenient in carrying
Operation of the product is simple ,low power consumption
SpO2 value display
Pulse rate value display, bar graph.displa
Pulse waveform display
The display mode can be changed
With backlight,and changeable backlight-time
A pulse rate sound indication
With measured data overruns limits and low-voltage alarm functionthe upper/down alarm range can be adjustable
With clock function
Battery capacity indication
Low-voltage indication: low-voltage indicator appears before working abnormally which is due to low-voltageand with alarm function
Connected with an external oximeter probe
Trend Graph/Table: Resolution from 1s, 5s, 10s,30s,1min,and so on. Storage of latest 96 hours trend data.
Alarm:Adjustable High and Low limits. Three level audible and visual alarm , latest 50 alarm information and waveform displays 8 seconds which 4 seconds are before a certain time when the alarm is turned on and 4 seconds are after appointed time when the alarm is turned on
Upload dataData stored in SD card can be uploaded to computer and analyzed by software “SpO2 Assistant”
Display Mode3.5″ Color TFT display
SpO2 Measuring Range0%100%, (the resolution is 1%).
Accuracy70%100%±2% ,Below 70% unspecified.
PR Measuring Range30bpm250bpm, (the resolution is 1bpm)
Accuracy±2bpm or ±2% (select larger)
Measurement Performance in Weak Filling Condition:SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ±4%, pulse rate error is ±2 bpm or ±2% (select larger).
Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1%.
Power Consumptionless than 300mA
Voltage: DC 5.0V
Power Supply: Built-in recharge Li-Polymer
Battery working hour:7 hours
Safety Type:Interior BatteryBF Type
Sell in standard
a user manual
a power adapter(Model:CMS0105
A power wire
a disk (PC software + a user manual)
An oximeter probe
Sell in addition
Weight: About136g(with battery)